FMU 860 PDFSeptember 15, 2020
Find great deals for Endress Hauser Prosonic FMU Ultrasonic Transmitter Measuring System. Shop with confidence on eBay!. matrix fmu – Download as PDF File .pdf), Text File .txt) or read online. FMU Ultrasonic Measurement Operating Instructions. FMU FMU FMU Endress + Hauser The Power of Know How.
|Published (Last):||18 March 2015|
|PDF File Size:||9.50 Mb|
|ePub File Size:||9.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
The counter can not be transmitted. The associated level is registeredautomatically.
Endresse Hauser FMU 860 Ultrasonic measurement
This shielding serves as a return cable and corresponds to the black wire of the unshortened cable. The counting factor Z1 in V1H5 is used for the optionally connected totaliser. The length unit is changed inmatrix field V8H3. Loosen the four cross-head screws of the operating plate.
A folded copy of the operatingmatrix is also found in the cover of the field housing. If a different linearisation is chosen, the manually entered characteristic remains storedin the Prosonic FMU, without being used. Commonscreened cable can be used. Source of supply Internet: Setting the operatingmodeNow enter a number in V8H0 for the mode: At the remaining instruments the terminating resistance must be switche off: Mounting screws and nuts supplied.
If the actualmaximum flowrate is entered in V2H7, then the value for the new width of the weir mustbe altered accordingly. Two Prosonic sensorsmeasure the level of the conical pile at separate points.
The totalizers switch in terms of the voluminar flow the counting unit V8H5 and the counting factors. Units of length may only be altered immediately after a reset of the transmitter After determing the length unit, this can only be altered when all other parametersare also changed.
The application parameters areselected at matrix position V0H3 for channel two V4H3 Application parameter 0, liquids Application parameter 1, liquids with rapid level changes Application parameter 2, fine-grained bulk solids Application parameter 3, coarse bulk solids Application parameter 4, bulk solids with rapid level changes. Touch an earthed object before handling the board. The linearizationtable is still in the memory, but no longer active.
The most important matrix fields for measured values are given in Table 3.
Open the operating plate4. A reset does not automatically set the display to a percentage! Mount the protective housing in a shaded area. FDU 83, 84, 85, In a protective 8660 IP 66 for wall or post mounting in the field or control room In a separate operating unit for control panel mounting, when the board withthe transmitter electronics can be mounted separately.
Endress Hauser Prosonic FMU 860 Ultrasonic Transmitter Measuring System
Setting of the analog outputs and the relays is described in chapters 6 and 7. The ffmu knock-outsin the connection area have to be removed for cable entry underside for 5 x Pg 16,4 x Pg Mountingmaterial screws or nuts for post mounting and the all weather protective cover isenclosed.
Theselevel applications are given in Appendix B. If strong sunlight is expected, then itis advisable to fit an all-weather protective cover obtainable as accessory. The level applications canbe selected in matrix position V0H3. To shorten the time taken for commissioning, for levelmeasurement, an application parameter can be selected which automaticallysets the ultrasonic measuring line to one of five different applications.
Endresse Hauser FMU Ultrasonic measurement | Vikat Ekinox
It is divided into the followingsections: This mode ensures that the product echo is evaluated even when the doubleecho is stronger. With Prosonic FMU a distinction can be made betweenchannels. Unregistered entries remain ineffective and the instrument willoperate with the old value.